OsteoCare™ one-piece SRBA is seated and fixed directly to the implant, while the two-piece SRBA abutment is seated and fixed to the implant internal hexagon by an abutment fastening screw, then the Polycarbonate ‘O’ Ring Housing can be processed into the denture either chair-side or at the dental laboratory.
The impression is taken by the indirect impression technique using an impression transfer. The one-piece or two-piece SRBA with the implant replica is used in the laboratory, to assist in the processing of the retentive anchor in the denture. The laboratory processes the Polycarbonate ‘O’ Ring Housing into the denture. At the stage of overdenture delivery, the one-piece SRBA abutment is seated and fixed to the implant internal hexagon directly or by a retaining screw if the two-piece SRBA is used.
Ball Attachments are designed to retain an overdenture to implants placed at angles from 0° to 45°. The one-piece SRBA is used to accommodate implant angulations from 0° to 15° with the two-piece SRBA already angulated from 15° to 45°.
Manufactured from Titanium alloy (Ti/6Al/4V) and has the same common hexagon interface as the Screw Retained Abutment (SRA). The coronal end of the SRBA is available with the OsteoCare™ Standard 2.00mm ball that allows connection to OsteoCare™ components
Secured by an Abutment Fastening Screw and tightened intraorally to 30N/cm and checked by the torque wrench.
|Screw Retained Ball Abutment||Product Code||Description|
|4.5mm||CO-BA4O-15 -> 45||2.0mm Ball Abutment 15º -> 45º (Two-Piece)|
|One-Piece Ball Abutment||Product Code||Description|
|3.75mm||CO-BA3OP-00S / 00L||2.0mm Ball Abutment 0º Short/Long (One-Piece)|
|4.50mm||CO-BA4OP-00S / 00L||2.0mm Ball Abutment 0º Short/Long (One-Piece)|
|5.00mm||CO-BA5OP-00S / 00L||2.0mm Ball Abutment 0º Short/Long (One-Piece)|